What are the signals for the RS-232 to RS-422 digital converter kit?
An RS-232 to RS-422 digital converter kit is required for all Turner devices utilizing a cable length greater than 10 meters. For communication with an X2 data logger, which is incompatible with the RS-422 sensor interface, the signal will need converted back to RS-232. The following wiring information outlines the power, ground and RS-422 signal output via a NexSens MCIL-8-FS flying lead cable.
What are the NexSens 8-Pin UW Connector signal connections?
If connecting a NexSens UW8 connector to an external device that is not an X-Series data logger, below is the pinout schematic of the NexSens UW8 connector, along with the pins associated with the communication, ground, and power signals for RS-232 sensors.
Why is a Turner C3/C6P Not Gathering Readings?
If a Turner C3/C6P Submersible Fluorometer is not gathering readings on a NexSens X-Series data logger after following the integration guide, here are a few common reasons that should be reviewed.